High-quality radar liquid level sensor. Not affected by changes in the properties of the medium, changes in the density of the measured medium or changes in the dielectric constant will not affect the measurement accuracy.
Radar level meter, radar is the acronym for Radio Detection and Ranging.
The radar liquid level sensor adopts the measurement method of the microwave pulse and can be normal within the industrial frequency band. The beam energy is low. It can be installed in various metal, non-metallic containers or pipes to conduct non-contact continuous measurements of the level of liquid, slurry, and particulate materials. It is suitable for occasions where dust, temperature, and pressure change greatly and inert gas and steam exist.
The radar liquid level meter is harmless to the human body and the environment and has the advantages of not being affected by the specific gravity of the medium, not being affected by the change of the dielectric constant, and not requiring on-site calibration. It is a good choice for both industrial needs and customers' economic benefits.
Radar level meter has become the mainstream product in the level measuring instrument market, which is mainly divided into radar level meter and guided wave radar level meter. If you are interested in the specific application of radar level meters, please feel free to contact us for more details.
|Product||Radar Liquid Level Sensor|
|Measure Range||0-30 meters|
Solid particles, powder
|Process Connection||Thread, Flange|
-0.1～4.0 MPa (flat flange); -0.1 ~ 0.3 MPa (universal flange)
|Signal Output||4-20mA (Two-wire/Four-wire)|
|Power Supply||2-wire（DC24V）/ 4-wire（DC24V /AC220V）|
Be installed in the diameter of the 1/4 or 1/6.
Note: The minimum distance from the tank
wall should be 200mm.
Note: ① datum
②The container center or axis of symmetry
The top conical tank level, can be installed at
the top of the tank is intermediate, can guarantee
the measurement to the conical bottom
A feed antenna to the vertical alignment surface.
If the surface is rough, stack angle must be used
to adjust the angle of cardan flange of the antenna
to the alignment surface.
(Due to the solid surface tilt will cause the echo attenuation, even Loss of signal.)